Dienmern Air Quality Detector PM2.5 portable Gas Analyzers & HCHO & TVOC &AQI air gas Meter Monitor Tester
Display method: 2.8” LCD screen display, 320 x 240 pixels
Atmospheric pressure: 86Kpa - 106Kpa
Detection method for PM: laser scattering
Sampling time: 1.5 seconds
Product Size: 155 x 87 x 35mm
Detection temperature: -10°C to 50°C; 14°F to 122°F
Relative humidity: 20% - 85%
Storage temperature: -10°C to 60°C; 14°F to 140°F
Concentration unit for HCHO and TVOC: mg/m³
Power source: Lithium battery with 1500 mAh capacity; 5V DC power charging via micro USB port
Product weight: 150g
This product is a multifunctional air quality detector that detects Formaldehyde (HCHO), Total Volatile Organic Compounds (TVOC), Particulate Matter <2.5 micron-sized particles (PM2.5/1.0/10), Temperature, and Humidity with clock and record function. As a scientific air quality detection device, it combines multiple air sensors with a built-in fan to allow real-time monitoring of formaldehyde (HCHO), total volatile organic compounds (TVOC), PM2.5/1.0/10, temperature, and humidity on its digital LCD display.
In many cities, they are polluted, so air quality needs to be verified. An ideal multi-function test tool for monitoring air quality and protecting your family's health if you are concerned about the air quality around you or your child. If it exceeds the standard, it will prompt the alarm! We also need to test indoor air quality, especially new renovated rooms. Don't forget, HCHO / TVOC formaldehyde is a healthy killer, especially for kids and babies. When we find unhealthy air quality (especially in a completely enclosed room such as a cool air-conditioned room or heating room. The air must be very unhealthy), we must open the door or window or use an air purifier to clean the air to make sure we have a Good healthy air environment.
1. The item must be placed indoors for more than an hour and then it
The actual data will be displayed. If you move randomly, the data will be inaccurate.
2. If you can ensure a clean environment, the goods can be used as calibration data and measures. If the item is low, the data may be inaccurate
1.Items Must Be Placed on The Indoor Static More Than An Hour Then It
Will Show The Actual Data. If Randomly Moved, The Data Will Be Inaccurate.
2.If You Can Ensure A Clean Environment, The Goods Can Be The Calibration Data, And Them Measure. If Items Are In Low Battery, The Data May Be Inaccurate
1 Q: What the test index are include for air quality monitor?
A : PM2.5: is main for smog(PM1.0 and PM10 are the same); HCHO: is for formaldehyde; TVOC: is main for benzene,etc Volatile organic compounds; AQI: is for Integrated air quality index.
2. Q: What is the safe range for formaldehyde (HCHO) and TVOC? A : HCHO≤ 0.10mg / m3 TVOC≤ 0.60mg / m3 According to ("indoor air quality standards" GB / T 18883-2002 and GB 50325-2001 Class II civil construction).
3 Q: What is the standard range for PM2.5? A: Good 0~12 Medium 12~35 safe value 35~70 Unhealthy more than 70
4 Q: Why do I need to calibrate(Just HCHO and TVOC need to calibration, PM2.5 and AQI no need)? A :Our products work in the form of outdoor formaldehyde-free air sampling as a benchmark, and then get indoor air formaldehyde content as a comparison, so the new machine products for a long time not used, the detection of environmental temperature changes, need to calibrated first and then tested, the value of this calibration for the subsequent detection of the accuracy of a great relevance.Calibration environment must avoid spices,perfumes, paint, cigarettes, air fresheners,alcohol and other chemical pollutants, so as not to damage the sensor.
5.Q: Why PM2.5 readings with the published value on the site is not the same? A : Because the location and conditions of the test is difficult to reconcile with the weather station in the city there are several sampling points, each sampling point measured data are also very different, the sampling point of the location is strictly required; height 3 to 30 meters, air circulation, can not be close to the chimney, furnace and other obvious sources of pollution. And there are no high-rise buildings, trees or other obstructions that can impede the flow of ambient air around the instrument.
6 Q: Why is the PM2.5 test value always changing? A : PM2.5 data are changing all the times, with the air, wind, humidity and other environmental factors change, in the room smoking, cooking fumes, car exhaust emissions, coal, chimney,furnace Such as pollution sources will change the PM2.5 value of this area, resulting in differences in the detection data.
7 Q: Why does the products make a sound when working? A : As PM2.5 detection need to collect a lot of mobile air, the inner fan needs strong operation to ensure accurate detection.
8 Q: How much range of detection of formaldehyde concentration detector,how many square meters? A : Test area≤ 50m3, room should be set 1~3 testing points; 50m3 ≤ Test area≤ 100m3, room should be set 3~5 testing points; ≥100m3, the points should more than 5 ponits. Testing points is on a diagonal or plum blossom evenly distributed, would more better. The height of the sampling point: in principle,consistent with the human breathing. Relative height of 0.5m ~ 1.5m between.
9 Q: which kinds of air index exceeded, would lead to Alarm sound? A: Just HCHO data more than 0.10mg / m3, would lead to Beep sound.