Langder High Sensitivity PM2.5 Dust Meter Air Analyzer
Large LED Screen with Bright Backlight, Convenient To Use
An Ideal Tool To Monitor The Formaldehyde/TVOC Concentration At Home
Use For Office, Inside A Car Or During Interior Decoration and Outdoor Environmental Testing
24 Hours of On-Going Real-Time Monitor Of Indoor Formaldehyde/TVOC
It iS Pocket-sized,Very Convenient to Carry and Measure The Air Auality Around You. This Device Can Measure PM ,HCHO,TVOC. Besides, This Detector Can Be Programmed and Revise the Interface According to Your Preference. We Will Provide The Programming Instruction.
Whats is Included:
1 Q: What the test index are include for air quality monitor? A PM2.5: is main for smog(PM1.0 and PM10 are the same); HCHO: is for formaldehyde; TVOC: is main for benzene,etc Volatile organic compounds; AQI: is for Integrated air quality index.
2. Q: What is the safe range for formaldehyde (HCHO) and TVOC? A : HCHO≤ 0.10mg / m3 TVOC≤ 0.60mg / m3 According to ("indoor air quality standards" GB / T 18883-2002 and GB 50325-2001 Class II civil construction).
3 Q: What is the standard range for PM2.5? Good 0~12 Medium 12~35 safe value 35~70 Unhealthy more than 70
4 Q: Why do I need to calibrate(Just HCHO and TVOC need to calibration, PM2.5 and AQI no need)? A :Our products work in the form of outdoor formaldehyde-free air sampling as a benchmark, and then get indoor air formaldehyde content as a comparison, so the new machine products for a long time not used, the detection of environmental temperature changes need to change First calibrated and then tested, the value of this calibration for the subsequent detection of the accuracy of a great relevance. Calibration environment must avoid spices,perfumes, paint, cigarettes, air fresheners, alcohol and other chemical pollutants, so as not to damage the sensor.
5.Q: Why PM2.5 readings with the published value on the site is not the same? A :Because the location and conditions of the test is difficult to reconcile with the weather station, the weather station in the city there are several sampling points, each sampling point measured data are also very different, the sampling point of the location is strictly required; height 3 to 30 meters, air circulation, can not be close to the chimney, furnace and other obvious sources of pollution. And there are no high-rise buildings, trees or other obstructions that can impede the flow of ambient air around the instrument.
6 Q: Why is the PM2.5 test value always changing? A : PM2.5 data are changing all the times, with the air, wind, humidity and other environmental factors change, in the room smoking, cooking fumes, car exhaust emissions, coal, chimney, furnace Such as pollution sources will change the PM2.5 value of this area, resulting in differences in the detection data.
7 Q: Why does the products make a sound when working? A : As PM2.5 detection need to collect a lot of mobile air, the inner fan needs strong operation to ensure accurate detection.
8 Q: How much range of detection of formaldehyde concentration detector,how many square meters? A : Test area≤ 50m3, room should be set 1~3 testing points; 50m3 ≤ Test area≤ 100m3, room should be set 3~5 testing points; ≥100m3, the points should more than 5 ponits. Testing points is on a diagonal or plum blossom evenly distributed, would more better. The height of the sampling point: in principle,consistent with the human breathing. Relative height of 0.5m ~ 1.5m between.
9 Q: which kinds of air index exceeded, would lead to Alarm sound? A: Just HCHO data more than 0.10mg / m3, would lead to Beep sound.